Leveraging the performance of the FAR-2xx8 and FAR-3xx0 Radars, the Wave Analyzer WV-100 allows real-time Finally, data sharing between ship and shore and between vessels is available for quick adjustment, economical route planning, and efﬁcient trafﬁc management around ports, where monitoring of sea conditions is required to guarantee the safety of all ships.
*As the Wave Analyzer is a computer-based program, it does not aﬀect the performance of the Radar system.
Possible to switch X-Band, S-Band and Solid State antennas according to weather conditions.
The information can be displayed on both the PC screen and the Radar screen.
It is the ﬁrst time that a Wave Analyzer incorporates an automatic adjustment function that frees the user from calibrating the Wave Analyzer following the installation.
In addition, the inﬂuence of a sudden sea condition change is minimal and does not affect the quality and accuracy of the wave information obtained.
Via serial transmission, data can be output to any external compatible system (Automated weather report system and Dynamic position system .etc).
Displays wave data, echo, and system information on a PC monitor.
The Color of the wave spectrum indicates the relative intensity of the wave spectrum display.
The absolute intensity of the radar reﬂection from the wave increases in order of dark blue, light blue, light green, dark green, orange, yellow, red.
The signiﬁcant wave height is a statistical value obtained by averaging the height of the one-third highest waves when observing the speciﬁed area.
Among the observed waves, the two most outstanding waves are selected and assigned as [1st Wave] and [2nd Wave] for reference. There is no distinction made between wind waves and swells for this selection.
Takes a screenshot.
|Wave Analyzer||MEASURING RANGE|
|Wave height||0.10 to 15.0 m|
|Wave period||4.00 to 16.00 s|
|Wave direction||2 max.|
|Bearing||-140 to +140° (from the bow)|
|Distance||150 to～ 1,200 m||SEA WEATHER CONDITIONS|
|Sea state code||3 minimum (wave height: 0.5 m or above)*|
|Beaufort scale||3 minimum (wind speed: 3.4 m/s or above)*|
|Sea Depth||0.5 times or more of the measured wavelength||MONITOR (USER SUPPLY)|
|LAN||2port -Ethernet 100Base-TX|
|RS-232C||1port (USB/RS-232C converter available)||Data sentence|
|PFEC (Output)||RAwv1, RAwv2||EQUIPMENT LIST|
|CD-ROM (Wave Analyzer and Operation Manual)
USB Protection Key